Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||20 October 2008|
|PDF File Size:||5.66 Mb|
|ePub File Size:||18.41 Mb|
|Price:||Free* [*Free Regsitration Required]|
Press Ok for launching measurement. If the probe has been just connected, the screen indicates the remaining time of warm up then the measurement screen appears, indicates the concentration and places it on a bargraph.
Kimo MP G Thermo Anemometer Manometer Air Velocity Pressure Sensor
In regards of Quality insurance kino, we recommend that the instruments are checked once a year. For others measurement modules, self-calibration is performed by disconnecting the two pressure inlets of the sensor, then by pressing Autoz key. On disabled mode, the probe is connected, the measurement is not carried out and the value is not displayed.
This datasets can’t be set using Datalogger Software. Memory capacity of the instrument is up to 8, points or 50 datasets.
Kimo MP 200 P User Manual
Click on Mesure to display values table of selected dataset. Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device.
Press 1x in order to select HOLD function: In regards of Quality insurance norms, we recommend that the instruments are checked once a year. Remove the front part at the back of the instrument. Probe display Available information are: Battery When battery indicator flashes it is recommended to change the batteries: ,imo on OK to validate.
When the measurement is fixed, press Min M200 to access to maximum and minimum values of the measurement.
To print the manual completely, please, download it. Configuration sub-function allows to: Planification must be made via the jp200. V — General informations Maintenance KIMO performs calibration, adjustment and maintenance of ,p200 your instruments to guarantee a constant level of quality of your measurements. Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed of several dated measuring points.
Click on Esc to return to previous screen.
On enabled mode, the probe is connected, the measurement is carried out and the value is displayed. Alarm threshold corresponds to the right value on the bargraph. This menu allows to view probe information plugged to C2, Module, C1 or wireless connections. This menu allows to view probe Probe display information plugged to C2, Module, C1 or wireless connections.
Select compensation It is possible to modify the value of the compensation in temperature. Select Start for dataset launching. With Pressure function, you can access to following sub-functions – Hold – see Air velocity – Config.
You can view each measuring points if you click on Visu.
Memory capacity of the instrument is up to 8, points or 50 datasets. By pressing ‘print’ button you will print only current page.
Air velocity with Pitot tube: Maintenance V — General informations Maintenance KIMO performs calibration, adjustment and maintenance of all your instruments iimo guarantee a constant level of quality of your measurements.
Go to Recording menu. Launch a planned dataset A planned dataset is composed of several locations. Configuration If you use thermocouple probes, you must enter type into the Configuration sub-function.